[Date Prev][Date Next][Thread Prev][Thread Next][Date Index][Thread Index]
>Also, we have a CF-3000 that is using more trichlor than our old CF-200.
>Any one else using one of these beasts?
I installed a CF-3000 here at VTA in November 1994 and had a problem of excess
solvent consumption. The problem turned out to be a leaking vapor accumulator
assembly. Lipsner-Smith took two resin containers and just stuck them
together without a gasket at the joint. When the cleaner was running you
could detect a strong solvent smell and feel the vapor leaking from the joint
where the two containers meet. Once this accumulator unit was replaced, I ran
a bead of sealer around the joint of the accumulator just to be sure no
solvent vapors leaked out. We run our CF-3000 approx. 10-15 hours a week and
go through approx. 15 gallons of solvent every 10 weeks.
Video Tape Associates