[Date Prev][Date Next][Thread Prev][Thread Next][Date Index][Thread Index]
Re: Panasonic Monitors
Larry Barth wrote:
>Are we the only facility in the world having any problems with the AT-H1905DP
> when the monitors are setup with the Panasonic probe,
>the numbers displayed when checked with a Minolta Color analyzer don't
>correspond to CIE specifications, although the color temperature of the
>display is correct. If the monitors are setup with the Minolta, the picture
>comes out warm. I know that the Minolta is correct, because all of our Sony
>monitors setup properly with the same device.
Are you using the RGB, or CIE mode on the Minolta? The RGB mode MUST be
properly setup and calibrated for each phosphor/CRT type (i.e. Sony
Broadcast, Panasonic, etc.) or it will not be accurate. Just a thought......
If you are in CIE mode and it doesn't match that's another story.......
Phil Mendelson philm at hollydig.com
"Animals are our friends...............
but they won't take us to the airport."